Rosemount Smart DP Transmitter exporter suppliers
Calibration of Smart Diferential Pressure Transmitter:
1. A 4-20mA fine-tuning is used to correct the D / A converter inside the transmitter because it does not involve a sensing component and requires no external pressure source.
2. Do a full fine tuning, so that 4-20mA, digital readings and the actual pressure signal is consistent, so the need for pressure signal source.
3. And finally adjust the range, by adjusting the analog output 4-20mA and external pressure signal source coincides with the role of the transmitter shell on the zero (Z), transfer range (R) switch the role of exactly the same.
How to ensure products no any problems ?
Rosemount pressure transmitter always taken of and other countries clients like it so much .
You may want to know how to promise products are high quanlity and original goods .
You can do not worry about this .our technical department will test each products for customers' order before shippment .Therefore , The clients can easy to purchase products from our company
More pressure transmittter customize welcome contact email@example.com 24h/7days service